Use of this information is subject to copyright laws and may require the permission of the owner of the information, as described in the ECHA Legal Notice.
EC number: -
CAS number: -
The determination of the flash point of 400160 was conducted according to EU method A.9, OPPTS 830.6315 and UN Manual Test Methods 32.4 in compliance with GLP standards. No reproducible flash point could be observed. After consideration of all measurements the minimum temperature for the flash point was used and the flash point will be stated as 178.5 °C at atmospheric pressure.
flash point tester following Pensky-Martens was used for the
determination. At first, the range of the flash point was determined
using a high heating range. Within this range, the determination of the
flash point was repeated five times. As no reproducible flash point
could be observed, a limit value for the flash point was determined
through a triplicate determination with the start temperature 190 °C.
400160 showed an irregular behaviour, showing flash points in the range
of 178.5 up to > 210 °C. Therefore, after conducting eight experiments,
the minimum temperature of 178.5 °C is considered as flash point at
inconsistencies in the measurements are attributed to the test item’s
properties: the mixture of different liquids are changing during the
heating phase of the instrument. Additionally, the test item might react
with the copper part of the vessel: a green colour of the liquid was
observed when emptying the test vessel after the performances. The flame
(when observed) was green, too.
Information on Registered Substances comes from registration dossiers which have been assigned a registration number. The assignment of a registration number does however not guarantee that the information in the dossier is correct or that the dossier is compliant with Regulation (EC) No 1907/2006 (the REACH Regulation). This information has not been reviewed or verified by the Agency or any other authority. The content is subject to change without prior notice.Reproduction or further distribution of this information may be subject to copyright protection. Use of the information without obtaining the permission from the owner(s) of the respective information might violate the rights of the owner.
Welcome to the ECHA website. This site is not fully supported in Internet Explorer 7 (and earlier versions). Please upgrade your Internet Explorer to a newer version.
Do not show this message again